|IGS-type acronym :||ACOR|
|Installed date :||1998-12-06|
|Decommissioned date :|
|City :||A CORUNA|
|Station status :||active (green)|
|Distance to Tide Gauge (m) :||2|
Sanchez Sobrino Jose A.
|IGS-like station form :|
|M||Material change or malfunction|
|RF||Reference Frame change|
Plots generated using software developed with the EPN team at ROB
|Observations & tracking performance||Satellite tracking|
The satellite tracking graph has been calculated for all stations for the first day of each month. If you wish to have a graph for another date, choose the 'Calculate a specific date' option.
Due to a fault of the receiver in ACOR station, data from DOYs 281 to 291 have been lost.
A new receiver Leica GR50 has been installed replacing the former GR10.
An update of ACOR site log has been made in M3G.
Former receiver Leica GRX1200 in ACOR station has been replaced today by a new Leica GR10.
Log file in M3G platform has been updated.
expected stoppage at the receiver:
And also the DOY 106 daily data is missing.
Due to a Leica GRX1200 bad behaviour in ACOR site, the tracking has been incorrectly from DOY 213-w to 232-l.