|IGS-type acronym :||CAS1|
|Installed date :||1994-01-01|
|Decommissioned date :|
|Station status :||active (green)|
|Distance to Tide Gauge (m) :||794|
|Station operator:||(GA) - Geoscience Australia|
|IGS-like station form :|
|M||Material change or malfunction|
|RF||Reference Frame change|
Plots generated using software developed with the EPN team at ROB
|Observations & tracking performance||Satellite tracking|
The satellite tracking graph has been calculated for all stations for the first day of each month. If you wish to have a graph for another date, choose the 'Calculate a specific date' option.
As reported in [IGSSTATION-7931] the receiver at CAS100ATA failed on 2020-04-04. A replacement receiver has been configured and operations returned to normal on 2020-05-22. Please not there may be a short delay in the RINEX files being pushed to IGS data centres over this weekend while we verify the quality.
The details of the change have been reported in the IGS site log manager.
We are having issue with the receiver at the IGS station CAS100ATA. We are trying to resolve it.
RINEX data is available till 4th of April,2020. There won’t be any RINEX files until further notice.
We will provide further information once the issue has been resolved.