|IGS-type acronym :||LAMP|
|Installed date :||1999-03-25|
|Decommissioned date :|
|Station status :||active (green)|
|Distance to Tide Gauge (m) :||130|
|IGS-like station form :|
|M||Material change or malfunction|
|RF||Reference Frame change|
Plots generated using software developed with the EPN team at ROB
|Observations & tracking performance||Satellite tracking|
Please be adviced that on DOY 2019/158 at approx 09:00 utc the ME only
firmware of the Leica GR10 receivers at LAMP (Lampedusa, Italy) and UCAG
(Univ. of Cagliari, Italy) stations was updated from 6.525 to 6.712.
Updated sitelogs have been submitted to EUREF.
On DOY 2018/206, LAMP (Lampedusa, Italy) Leica GR10 firmware was updated
from 4.11/6.523 to 4.20/6.524.
Updated sitelog was submitted to EUREF.
On doy 2017/269 LAMP (Lampedusa, Italy) station resumed operations.
The tracking problem was fixed replacing antenna and cable.
The Leica AR25 antenna was replaced by Leica AR20.
Updated sitelog has been submitted to EUREF.
On doy 2017/255 LAMP (Lampedusa, Italy) station stopped to track all
GNSS satellites for still unknown reason. The reciver is accessible from
remote, so we think it could be a cable/antenna problem.
Investigation is in progress.
As you probabilly already noted, starting from doy 2017/171, data from
LAMP (Lampedusa, Italy) station are missing.
We lost link to the station so we need to go there to restore it and
unfortunately this needs some time.
On doy 2013/320 LAMP station (Lampedusa, Italy) stopped to send data. The
reason of the outage is under investigation, I will keep you informed.
[EUREFmail-7223] - 2013-11-27: On doy 2013/330 data flow from LAMP (Lampedusa, Italy) resumed. We are checking the quality of the data acquired.