|IGS-type acronym :||POHN|
|Installed date :||2003-01-10|
|Decommissioned date :|
|Country :||FEDERATED STATES OF MICRONESIA|
|Station status :||No data for 30 days (orange)|
|Distance to Tide Gauge (m) :||2539|
|IGS-like station form :|
|M||Material change or malfunction|
|RF||Reference Frame change|
Plots generated using software developed with the EPN team at ROB
|Observations & tracking performance||Satellite tracking|
The IGS station POHN00FSM is currently offline. We are currently troubleshooting to determine the cause, and aim to have the site back online. RINEX and real time data will be unavailable until the issue is resolved.
The antenna at the IGS site POHN in Pohnpei (FSM) has failed. We are currently looking into options of getting a replacement antenna to the site. Given our current work program it is likely that this problem will not be resolved until January 2018.
In the meantime no real-time streams or RINEX products will be available.
We have recently been alerted to poor quality data coming from the IGS station POHN. The problem is believed to be hardware related. As a result we are temporarily suspending the distribution of data from this station.
Further notification will be given when the problem has been resolved.
This email is to report an equipment failure and subsequent data outage at the IGS tracking site POHN. The data outage began on day of year 201. We are trying to get in touch with our local contacts to assess the situation.
The site POHN is currently experiencing a data outage, the outage began on day of year 102, and is most likely a result of equipment malfunction. We are currently looking into the issue and further notification will be given when the site is operational again.
Please be advised that the Trimble Chokering (TRM59800.00 NONE) at the IGS site POHN has been replaced. The updated log is available on our FTP server. The data outage from DOY 14064 – 14084 will be permanent.
Please be advised that the Trimble Chokering (TRM59800.00 NONE) at the IGS site POHN has seen a slow degradation in tracking of L2 over recent months to the point where the data has become unusable. The antenna was removed on 2014-03-06 and a replacement has been sent.
Further notification will be given when the antenna has been replaced.