|IGS-type acronym :||CHTG|
|Installed date :||2015-02-20|
|Decommissioned date :|
|Station status :||active (green)|
|Station operator:||(LIENSs - OASU) - LIENSs - OASU|
|IGS-like station form :|
|M||Material change or malfunction|
|RF||Reference Frame change|
Plots generated using software developed with the EPN team at ROB
|Observations & tracking performance||Satellite tracking|
The satellite tracking graph has been calculated for all stations for the first day of each month. If you wish to have a graph for another date, choose the 'Calculate a specific date' option.
Intervention by the SHOM to change the acquisition central of the tide gauge. Before it was an Elta system, now it is a Campbell. Also, installation of the meteorological sensor and the camera
Following message 2020-11-03, T.Donal explains that the increase in maritime traffic and ship mooring nearby CHTG GNSS station creates more masks that affect the GNSS signal reception. E.Poirier (ULR)
M.Gravelle and T.Donal spotted anomalies in CHTG E,N,U series. This may be due to a high number of cycle slips since may-2018 especially in the norhtern sector of the antenna. Investigation is started to find out the reason for this cycle slips. Thre could be an obstruction in the North of the antenna. E.Poirier (ULR)
Dataflow for the CHTG station has restarted since session 2018-226t. Missing data are definitely lost.
Firmware is updated to version 3.6.13 Feb,12,2018, and the sitelog is updated
Due to a technical problem, the CHTG station is unavailable. The delay to recover a normal trafic is unknown.