|IGS-type acronym :||SFER|
|Installed date :||1995-12-18|
|Decommissioned date :|
|Station status :||active (green)|
|Distance to Tide Gauge (m) :||10311|
|IGS-like station form :|
|M||Material change or malfunction|
|RF||Reference Frame change|
Plots generated using software developed with the EPN team at ROB
|Observations & tracking performance||Satellite tracking|
Due to technical failure in the storage card of our equipment,
daily RINEX 2
On Thursday 17th January, we replace following components:
- Antenna LEICA AR25 replacing previous LEICA AR20
- Antenna cable type LMR400
Besides the New POC of the Station is Manuel Angel Sánchez Piedra
A mistake has been appreciated on SFER site log. Current receiver firmware
On Wednesday 20th of june SFER Leica GR-25 receptor firmware was updated
from version 4.11.606 to version 4.20.232. The station is working again
using internal oscillator due to technical issues with H-maser.
Due to issues with SFER receptor the station will be offline until technical
service check it.
This morning SFER Leica GR-25 receptor firmware was updated to version
4.11.606. We hope this solve the problems we have with the station.