|IGS-type acronym :||SFER|
|Installed date :||1995-12-18|
|Decommissioned date :|
|Station status :||active (green)|
|Distance to Tide Gauge (m) :||11094|
|Station operator:||(RIOA) - Real Instituto y Observatorio de la Armada|
|IGS-like station form :|
|M||Material change or malfunction|
|RF||Reference Frame change|
Plots generated using software developed with the EPN team at ROB
|Observations & tracking performance||Satellite tracking|
On Wednesday 17th of June, the firmware of SFER Leica GR25 receiver has been
update from 4.20/6.713 to 4.31/6.713. Log file has also been updated.
On Thursday 11th of June, the ME firmware of SFER Leica GR25 receiver has
been update to 6.713 (4.20.232/6.713).
Log file has also been updated.
Today, we have stopped submitting RINEX2 data for SFER station.
RINEX3 data continues to be submitted as usual.
Due to technical failure in the storage card of our equipment,
daily RINEX 2
On Thursday 17th January, we replace following components:
- Antenna LEICA AR25 replacing previous LEICA AR20
- Antenna cable type LMR400
Besides the New POC of the Station is Manuel Angel Sánchez Piedra
A mistake has been appreciated on SFER site log. Current receiver firmware
On Wednesday 20th of june SFER Leica GR-25 receptor firmware was updated
from version 4.11.606 to version 4.20.232. The station is working again
using internal oscillator due to technical issues with H-maser.
Due to issues with SFER receptor the station will be offline until technical
service check it.
This morning SFER Leica GR-25 receptor firmware was updated to version
4.11.606. We hope this solve the problems we have with the station.